Title & Document Type: A 50 MHz Programmable Pulse Generator This manual may contain references to HP or Hewlett-Packard. Hp A Pdf User Manuals. View online or download Hp A Operating, Programming And Servicing Manual. View and Download HP A operating, programming and servicing manual online. 50 MHz programmable. A Pulse Generator pdf manual download.
|Published (Last):||25 September 2016|
|PDF File Size:||18.61 Mb|
|ePub File Size:||13.4 Mb|
|Price:||Free* [*Free Regsitration Required]|
These components are summarised in Table Free Running Signature Analysis To carry out signature analysis the microprocessor must be set to a free-running routine in which it increments through all addresses continuously.
Switch on the HP A and check that error number E52 is displayed. Home Questions Tags Users Unanswered. When a line functions as an output, it will sink up to 48 mA in the low state and approximately 0. Remove the heatsink s.
Contact the seller – opens in a new window or tab and request a shipping method to your location. It is also used as a burst generator.
Connect the equipment as shown in the setup figure 2. Verify that the High-level output can be varied between – -8 V and —8 V while the —2 V level remains unchanged as shown below. Hewlett-Packard shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance or use of this material.
Recommended Test Equipment 3- 1. Instrument Overview This chapter deals with the overall instrument and is intended to help you to isolate a fault at a functional level.
It is also possible to read all the instrument settings in one go using the CST mnemonic. Same as triggered mode except pulses are output for the duration of the gated signal.
From London Hackspace Wiki. Verify that voltage at U pins 1,4 and 5 are at virtual ground min.
HP Hewlett Packard 50 MHz Pulse Generator 8112A
Delay Generator Table Width Generator Operation The offset signal is applied to the non-inverting input at the junction of CR, CR to ensure a constant reference point. Test Preperetipn Refer to Chapter Manuwl two basic ranges, time and frequency, are further divided if necessary by switching divider circuits 6 X decade counters into operation under control of the range setting. Delay Double Pulse I can understand if English and Engineering are not your 1st languages just ask my wife when I speak engineering talk.
Check that all inter-connecting co-axial and flat cables are properly connected.
The HP A contains static- sensitive devices. Adjustment Procedures – Changeable Components HB-IB Connector 4- 1.
HP A Compatible Pulse Generator – GPH | Marvin Test Solutions, Inc.
For example, STO 3 stores the current setting in location 3. If a modulated output is required, select the required modulation using the [CTRL] key. Set the attenuation factor to and its offset to 0 V.
Puise Width Performance Test 1. Output quiescent until triggered by an internal, external, GPIB or manual trigger, then generates one cycle at manuxl period rate. Range Switches The ramp timing capacitance is selected from range latch U Principle of DAC Operation Insfrument Overview and Troubleshooting Guide All the front panel LEDs should light momentarily.
HP 8112A Operating, Programming And Servicing Manual
Adjust A1R for best pulse droop mwnual all three transition modes. Set up the DVM as follows: Typical signals in External Burst mode Interest will be charged to your account from the purchase date if the balance is not paid in full within 6 months.
To set this jumper proceed as follows: The timing ICs used in the HP A are programmable timers which can be used to produce repetition rates, pulse widths and delay times by utilizing a variety of trigger and gate mode inputs. Ensure the Line Voltage Selector switches are set properly for the power source to be used.
The output from U19 therefore represents the state of the key joining the vertical ph path addressed by A3-A5 and the horizontal signal path addressed by A0-A2.
Re-enable the output signal.
If re-adjustment is necessary, repeat Fixed, Linear and Gauss adjustments as required. Period Generator output signals These bus management lines are used to control the flow of data on the HP-IB data lines.