The ELx continues our tradition of offering the tightest specifications of any microplate reader. Visit BioTek to learn more about our precision microplate. ELx Operator’s Manual. Revision History. Rev Date Changes. A. 1/96 First Release. B. 5/96 Revised reader specifications. Added Scanning. View and Download Biotek ELx operator’s manual online. Absorbance Microplate Reader. ELx Laboratory Equipment pdf manual download. Also for.
|Published (Last):||24 August 2017|
|PDF File Size:||9.6 Mb|
|ePub File Size:||4.70 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page To define a cutoff nanual for Positive and Negative calls: Recommendations For Optimum Performance Remain aware of the total number of controls, standards, and blanks that were originally mapped in each individual assay while mapping for the Panel assay.
Biotek ELx808 Manuals
To start running a reader program, press the Read key. The ELx will automatically perform a System Test.
Assay Detail Report By pressing ‘print’ button you will print only current page. A failed fuse is usually an indication of another problem that a new fuse is not likely to remedy. Remove the top shroud of the instrument see instructions in Install the Filter Wheel Page 35 Page 36 – 4: Page Failed configuration checksum test.
Perform The Test The well that shows the most variation for each concentration is selected for data reduction. Prepare Test Solutions Plate screen to select one wavelength.
Error Codes (general) – Biotek ELx Operator’s Manual [Page ]
Liquid Tests Liquid Tests Prepare Test Solutions BioTek offers a dye solution PN25 mL; ormL that may be used in manial stock solution formulation for Liquid Tests 1 and 2, or, if you prefer, you may use the dye solution described below. Defective motor drive circuit or volt power problem. The software will automatically increment each subsequent sample identification number by 1.
Page Chapter 6: Before Repackaging The Instrument This shipping system was designed to be used no more than five times.
The Louisville reference center recommends the following positive discrimination criteria according to the recommendation of the 2nd International Anticardiolipin Workshop.
Average temperature of the four zones of the incubator at the time the system manhal is run.
Inspect the glass filters for speckled surfaces or a halo effect. Optional Connect Printer 6: Edited and formatted text.
Microplate Readers: ELx Absorbance Microplate Reader – Overview
To avoid system instability, make sure the printer and reader are turned OFF before connecting them. Package Contents Package contents and part numbers are subject to change. Enter text from picture: Perform The Tests After the plate has been read, all requested reports will be generated. Print the ten sets of raw data or export them to an Excel spreadsheet using Gen5.
Perform Liquid Test 3 “uv” Models Only Roller surface and hook-in bracket BioTek Instruments, Inc. Greyzone Entry El8x08, for example, POS is selected as shown in the above screen, calls will be assigned according to the following equations SMP represents the sample wells: Transformation Formulas Divide all ODs on the plate by 2 and multiply by Reconfigure Or Replace Fuses Page of Go. Page 8 System Test