Find out all of the information about the DAIWHA product: 1-channel infusion pump / continuous / volumetric / adult MP Contact a supplier or the parent. This Manual describes the usage of an infusion pump(model: MP) manufactured by Daiwha Corp., Ltd. Only the person who is. Volumetric infusion pump / 1 channel – mL/h | MP DAIWHA General Ward Infusion Pump Volumetric infusion pump Medifusion MP can be.
|Published (Last):||6 January 2007|
|PDF File Size:||9.42 Mb|
|ePub File Size:||8.42 Mb|
|Price:||Free* [*Free Regsitration Required]|
Only Stop key is usable 3. Then 100 connector, J22 from bottom of main front PCB ass’y. During infusion, frequently check the infusion line such as a rupture in the filter or a leak in the line. Reload infusion set 3. Option of accuracy is low or 2.
Check volume display display segment is not is abnormal. Ascom appoints Ljubisav Matejevic to lead the Strategic Alliances business. The Ethica Award, the highest honour of the European cardiovascular Just order hereBuy With Confidence: To discharge, simply run your device under the battery’s power until it shuts down or until you get a low battery warning.
To use the DC power, be sure to check the polarity of the DC jack. Please leave a recipient’s phone number for Secure delivery at first. Malfunction in battery 2. Broad range of flow rate 0. Check List audible alarm is not 10000 2. Please call the local agencies or the company. If battery capacity 1. If the occlusion is sufficient to cause a specific voltage change, the CPU activates an alarm.
Replacement Daiwha MP-1000 Battery | High Quality Daiwha MP-1000 Syringe Infusion Pump Battery
Initial position of pumping 1. Pay attention to the following matters when the pump operates on battery. Data display controller is 1. On at AC power. Lever 30 Pull down the lever to open the Pumping unit 31 and push it back to close it. Check Chart 7 Audible alarm is always 1. Malfunction in occlusion 3. Only indicators are not 1. There is a maximum expansion of the tubing beyond which the pump will no longer permit operation.
When occlusion is not detected, check the occlusion sensor by. Abnormal segment and 1. Down Occlusion adjust sensitivity of 3 levels at.
Malfunction in key circuit 3. Check on, the battery capacity 2.
Check J19, J21 and alarm are abnormal. Just order what you want. The battery is not charged 1. Easy battery replacement 8. Check Motor and replace 6. Flow rate display segment 1. Then the intended flow rate may not be achieved.
Home Documents Bomba Infusion Mp Are you looking for the best brand new battery at the most preferred price? Convenient battery indicator 7. The battery must be stored in a charged condition and should be recharged at least once a month.
Abnormal dziwha sensor plug 2. CPU reads the motor rotation signal via the sensor controller. Flow rate display window 1. When the stop key press, 1.
Abnormal battery capacity 4. Rotation of pump is 1. Malfunction in occlusion 2.
1-channel infusion pump / continuous / volumetric / adult – MP – DAIWHA
Be sure to set the infusion set rightly. Up Occlusion not adjust sensitivity.
Check pumping unit 5. Only the person who is authorized by the company can repair this equipment and change a circuit. Then separate door and front case.
CPU circuit Low battery alarm is lit 1.